Clinical and echocardiographic outcomes after implantation of the Trifecta aortic bioprosthesis: an initial single-centre experience.
OPEN Interactive cardiovascular and thoracic surgery | 20 Nov 2012
AM Dell'aquila, D Schlarb, SR Schneider, JR Sindermann, A Hoffmeier, G Kaleschke, S Martens and A Rukosujew
OBJECTIVESThe Trifecta valve (St. Jude Medical) was introduced into clinical practice as a tri-leaflet stented pericardial valve designed for supra-annular placement in the aortic position. The present study aims to evaluate the preliminary results with this new bioprosthesis.METHODSSeventy patients underwent aortic valve replacement (AVR) with the Trifecta valve between August 2010 and December 2011. Thirty-three patients were male and 37 were female (52.9%). Mean age was 74.65 ± 7.63 (range 47-90 years). Prevalent cause of AVR was aortic stenosis in 64 (91.43%) patients. The mean preoperative pressure gradient was 50 ± 17 (range 20-84 mmHg), and the mean aortic valve area was 0.77 ± 0.33. Five (7.14%) patients were operated on due to aortic valve endocarditis. One patient was operated on due to isolated, severe aortic insufficiency. All patients were in New York Heart Association functional class III or IV. Twenty-eight (40%) patients underwent concomitant procedures.RESULTSConcomitant procedures were coronary artery bypass grafting (n = 25), mitral valve replacement (n = 1), ablation of atrial fibrillation (n = 1) and septal myomectomy (n = 1). There were no intraoperative deaths. The 30-day in-hospital mortality was 2.85% (2 of 70). One late death occurred during the in-hospital stay due to a multiorgan failure on postoperative day 60. There were 2 (2.85%) perioperative strokes. Mean pressure gradient decreased significantly from a preoperative value of 50 ± 17 mmHg to an intraoperative gradient of 9 ± 4 mmHg (Table 3). The mean gradients were 14, 11, 11, 8 and 6 mmHg for the 19, 21, 23, 25 and 27 mm valve size, respectively. No prosthesis dislocation, endocarditis, valve thrombosis or relevant aortic regurgitation was observed at discharge.CONCLUSIONSThe initial experience with the Trifecta valve bioprosthesis shows excellent outcomes with favourable early haemodynamics. Further studies with longer follow-up are needed to confirm those preliminary results.
* Data courtesy of Altmetric.com